Add : No.9, Changchun Rd, High-tech Development Zone, Zhengzhou, China 450001
Could I cut short probe when installation? Beucause I offer wrong tank data when ordering.
Time:2018-08-07 10:32:11 Author:yisainuo2018
It forbidden. The tank level gauge probe is manufactured strictly base on the tank size and site condition, the user who offer wrong tank data will find probe too long or too short when install, but the probe can not be cut short after delivery out of factory.
It will bring two serious result and dangerous. Firstly it will enlarge measure error, highly lower precision; secondly construct probe again will bring huge potential dangerous on safey working. The probe is working inside tank, oil penetrate into stainless steel pipe because of poor seament.